 Info Sheets 
                        
 Out side 
    

    
Result : Searchterm 'Real' found in 4 terms [] and 43 definitions []
 1  5 (of 47) nextResult Pages : [1] [2 3 4 5 6 7 8 9 10]     Searchterm 'Real' was also found in the following services:     
  
Real  
 

A real number is a component of a complex number. In MRI, complex data are used for example in the Fourier transforms. 
     • Share the entry 'Real':    • View the NEWS results for 'Real' (16).
    Further Reading:  Basics:


 
   Searchterm 'Real' was also found in the following services:     
  
 

Immediate display of reconstructed images during the scan time. Also called online display. 
    • View the DATABASE results for 'Real Time' (5).
   • View the NEWS results for 'Real Time' (6).
    Further Reading:   Basics:


News & More:
 
 
        Searchterm 'Real' was also found in the following services:     
  
 

Change of the measurement parameters in real time. 
     
   Searchterm 'Real' was also found in the following services:     
  
 

The partial Fourier technique is a modification of the Fourier transformation imaging method used in MRI in which the symmetry of the raw data in kspace is used to reduce the data acquisition time by acquiring only a part of kspace data.
The symmetry in kspace is a basic property of Fourier transformation and is called Hermitian symmetry. Thus, for the case of a real valued function g, the data on one half of kspace can be used to generate the data on the other half.
Utilization of this symmetry to reduce the acquisition time depends on whether the MRI problem obeys the assumption made above, i.e. that the function being characterized is real.
The function imaged in MRI is the distribution of transverse magnetization Mxy, which is a vector quantity having a magnitude, and a direction in the transverse plane. A convenient mathematical notation is to use a complex number to denote a vector quantity such as the transverse magnetization, by assigning the x'component of the magnetization to the real part of the number and the y'component to the imaginary part. (Sometimes, this mathematical convenience is stretched somewhat, and the magnetization is described as having a real component and an imaginary component. Physically, the x' and y' components of Mxy are equally 'real' in the tangible sense.)
Thus, from the known symmetry properties for the Fourier transformation of a real valued function, if the transverse magnetization is entirely in the x'component (i.e. the y'component is zero), then an image can be formed from the data for only half of kspace (ignoring the effects of the imaging gradients, e.g. the readout and phase encoding gradients).
The conditions under which Hermitian symmetry holds and the corrections that must be applied when the assumption is not strictly obeyed must be considered.
There are a variety of factors that can change the phase of the transverse magnetization:
Off resonance (e.g. chemical shift and magnetic field inhomogeneity cause local phase shifts in gradient echo pulse sequences. This is less of a problem in spin echo pulse sequences.
Flow and motion in the presence of gradients also cause phase shifts. Effects of the radio frequency RF pulses can also cause phase shifts in the image, especially when different coils are used to transmit and receive.
Only, if one can assume that the phase shifts are slowly varying across the object (i.e. not completely independent in each pixel) significant benefits can still be obtained. To avoid problems due to slowly varying phase shifts in the object, more than one half of kspace must be covered. Thus, both sides of kspace are measured in a low spatial frequency range while at higher frequencies they are measured only on one side. The fully sampled low frequency portion is used to characterize (and correct for) the slowly varying phase shifts.
Several reconstruction algorithms are available to achieve this. The size of the fully sampled region is dependent on the spatial frequency content of the phase shifts.
The partial Fourier method can be employed to reduce the number of phase encoding values used and therefore to reduce the scan time. This method is sometimes called halfNEX, 3/4NEX imaging, etc. (NEX/NSA).
The scan time reduction comes at the expense of signal to noise ratio (SNR).
Partial kspace coverage is also useable in the readout direction. To accomplish this, the dephasing gradient in the readout direction is reduced, and the duration of the readout gradient and the data acquisition window are shortened.
This is often used in gradient echo imaging to reduce the echo time (TE). The benefit is at the expense in SNR, although this may be partly offset by the reduced echo time. Partial Fourier imaging should not be used when phase information is eligible, as in phase contrast angiography.
See also acronyms for 'partial Fourier techniques' from different manufacturers. 
    • View the DATABASE results for 'Partial Fourier Technique' (6).
    
    
 
   

 
 Look Ups 
 